22 May 2003 HumaLyzer 3000 filtration photometer one ray with constantly spinning filtration wheel Monochromatic or bichromatic scanning 8 filtration jobs 330 to 670 nm start and also by kept selection Absorbance one Standard Differential samples element means Differential samples multiple requirement means (up to 7 criteria) Multi Standard per cent Abs (up to 7 standards) Kinetic setting (repeatedly, or simultaneously (Batch)) By aspect or by traditional
Stations Source of Radiation Selection of Wavelength Filter sort Wavelength precision filtration Location Filter option Wavelengths one half Bandwidth 1/100 Bandwidth False vibrant Fuel proportion Cuvette Supplied kind Material Geometry Illuminated quantity Minimum study levels Aspiration/Purge device Cuvette owner alarm alert running and screen show kind level of screen Absorbance focus energizing Results Zero Compensation Range alert Outputs Parallel Serial data-input Spectrophotometric Inaccuracy Flow-through
Fixed opportunity energizing By aspect or by regular 44 Pre-Configured, 76 start Tungsten Halogen, 10 Watt, with automatic light saver By filtration 4-cavity disturbance, long-life ion beam-assisted deposition +/- 3 nm After sample (heat absorbing filter before sample) auto by pc software or via keyboard 340, 405, 505, 545, 580, 630 nm offered regular other/additional filter systems elective 50%. When full, push INSERT to go back on previous prompt. Brand-new water prices need to be look over as described above in Read liquids Reference. 2.1.2 Lamp heat and Light Saver The light must limber up before every indication are taken. Once the light is actually aroused the warm-up opportunity begins and consistently count straight down. If a test is chosen rapidly, the device will stop till the warm-up the years have elapsed. The content light heat:XX Secs try displayed and the time XX will depend down seriously to zero. Whenever the warm-up is complete the tool will continue. If tool is not utilized in roughly 15 minutes, the light will immediately be deterred to extend this service membership life of the lamp. Choose a test or function, or click LAMP to make the light on. 2.1.3 Additional harbors An external printer may be attached to the synchronous port on the back in the unit. Any IBM appropriate printer and regular synchronous wire can be utilized using the parallel interface. The serial port on the HumaLyzer 3000 are MultiDirectional (transmits and get information). The receiving equipment may momentarily quit the indication from the tool. It is going to resume automatically if the receiving product is prepared. If the receiving device does not have a large enough buffer, the HumaLyzer 3000 times out and the Ext Printer Not Ready, Retry Y/N? error message appears on the display. This mistake information means any receiving tool (printer, Computer, exterior serial or parallel harbors). 2.1.4 Serial interface through the use of a general RS232 wire (Null Modem Serial Cable), a personal computer with a serial slot can be connected. The information format is actually 9600 baud, 8 parts, 1 stop little, no parity. Get hold of your supplier to get a serial printer cable.
15 Real HumaLyzer 3000 Consumer Handbook
2.1.5 Computer Connection The serial interface pinout has become reconfigured to permit drive connection to a PC. The production might be viewed on a PC making use of a program such as for example HyperTerminal when the marketing and sales communications method when you look at the system is set-to the HumaLyzer 3000 information style. 3 TD (SEND INFO output) 5 GND (ALERT SURFACE) 8 CTS (CLEAR TO DELIVER input) consult 2.1.6 to find out more. 2.1.6 Mistake Messages when using the match or Serial slots In the event the exterior printer or PC was allowed, in addition to printer or PC is connected it is off-line, one https://datingmentor.org/escort/mcallen/ benefits only once the instrument attempts to speak when in a mode. The instrument prevents and exhibits serial radio not prepared before problem is corrected. In the event that the outside printer or PC buffer becomes complete (in other words. the instrument have delivered the maximum amount of data once the printers mind can take), the instruments show will encourage to Retry. Click YES to keep to output after the printer or Computer keeps emptied their buffer, or NO to show down telecommunications to external harbors. 2.1.7 Units of description in every settings except Absorbance setting, a choice of selecting products of measurement is provided with. Products are given to label the measured concentrations, but have no bearing regarding real computation. The screen demonstrates: